versión On-line ISSN 2309-8309
versión impresa ISSN 1681-150X
SA orthop. j. vol.9 no.3 Pretoria ene. 2010
J WaltersI; M SolomonsII; S RocheIII
IFC(Ortho)SA. Department of Orthopaedic Surgery, University of Cape Town
IIFC(Ortho)SA. Department of Orthopaedic Surgery, University of Cape Town
IIIFC(Ortho)SA. Department of Orthopaedic Surgery, University of Cape Town
Recognition of unfavourable outcomes in displaced fractures of the clavicle in its middle third has stimulated global interest in the fixation of these fractures. The design of fixation devices is dependent on in-depth knowledge of the dimensions and morphology of the clavicle in order to optimise the configuration and fixation points of such a device.
In this study 100 dry clavicles were obtained from the Anatomy Department for assessment of the dimensions, angular deviations and gross morphology. The clavicle could be divided into distinct regions according to its muscle and ligament attachments and the cross-sectional morphology was typified for each region. The physical data presented can be utilised to inform fracture or osteotomy fixation-device design.
Clavicle fractures are common and occur most frequently (80-85%) in the middle third. Although usually treated conservatively, there is increasing recognition of the morbidity associated with displaced fractures, and also of the increased rate of non-union of fractures in this zone. The improved benefits derived from fixation have resulted in renewed interest in fixation of clavicle fractures. While plating and nailing have been popularised, an external fixation device can also be effectively employed for the midthird fractures as reported by Schuind,1 Demiralp,2 Tomic3 and others. This renewed interest in fixation has stimulated closer scrutiny of the anatomy of the clavicle
The design of fixation devices depends to a large extent on the anatomical and physical characteristics of this bone. In order to get data on the normal variation of the dimensions of the clavicle to assist with design, a search of the literature was undertaken. Most studies reporting the physical characteristics and appearance of the clavicle are published in the forensic medicine literature in order to capture data to assist with gender identification and size (length) of the deceased individual.4,5
Few anthropometric studies address this bone from a biomechanical perspective. Ahmad et al examined the morphometric properties of the clavicle in 88 subjects in the UK, by using a 3D reconstruction of CT images6 with particular reference to the diameter of the intramedullary canal, and Gumina et al, in a study of the prevalence of a coraco-clavicular joint, reported only on the length of 509 dry clavicles of Italian origin as measured on radiographs.7 In a study of 100 pairs of clavicles by Huang et al, they record the medial and lateral angle and the clavicular bow angle with specific relevance to screw and plate fixation.8 A study by Daruwalla et al9 (UK) described a technique by which they examined 15 clavicles by 3D CT reconstruction. They concluded that the use of CT and 3D statistical shape analysis is advantageous because it rules out observer variability and uses 'live' clavicles, the dimensions of which have been shown to differ appreciably from dry bone.10
To date, no published study on the morphology and dimensions has been adequate enough to permit the calculation of the forces that are transmitted through the clavicle during activity. To obtain the precise data that we required to investigate the range in size and morphological features of the human clavicle we undertook a survey of dry cadaver clavicles which had been sourced from the Anatomy Department.
Due to cost constraints we were not able to gain access to radiographic images for the morphological and dimensional assessment.
The purpose for undertaking this study was to obtain baseline data on length, cross-sectional morphology, medial and lateral angles and the clavicular bow in the transverse plane. Such information can be used for the theoretical calculation of the forces and stresses that are transmitted through the clavicle during low grade activity, such as elevating or abducting the arm, and to assist with the design of clavicle fracture fixation devices so that they are able to withstand the applied forces.
Materials and methods
One hundred clavicles collected between 1987 and 1994 were obtained from the Anatomy Department for analysis. Forty-five were left-sided and 55 right-sided. Age, race and gender differentiation were unknown.
Initial scrutiny revealed that the clavicle could be divided into five distinct zones as determined by the surface markings of its muscle and ligament attachments (as demonstrated in Figure 1) and the cross-sectional morphology. The reason for considering these landmarks was based on the theoretical assumption that fixation points on the clavicle for a fixation device would be best achieved in such a way that the muscles or ligaments would not be affected adversely by the fixation device.
The clavicle was divided at the following sites as shown in Figure 2:
Point A and F represent the medial and lateral ends of the clavicle.
Point B which represents the lateral most extent of the insertion of the sternocleidomastoid muscle superiorly, and sternoclavicular ligament inferiorly; A/B represents zone I and B/C represents zone II.
Point C is the junction of the medial and middle thirds of the clavicle, so that A/C represents the medial third.
Point D is the junction of the middle and lateral thirds of the clavicle so that C/D represents the middle third of the clavicle. Inferiorly subclavius attaches roughly between points C/D which represents zone III.
Point E is the midpoint of the outer third where a significant change of cross-sectional profile occurs between zones IV and V.
D/F represents the outer third of the clavicle; D/E represents zone IV, and E/F is zone V.
The following measurements were made:
Using Vernier callipers, the overall length of the clavicle and length of each zone were measured. The cross-sectional dimensions at the designated sites were measured perpendicular to the bone in the midsection of each zone.
2. Cross-sectional morphology
The cross-sectional morphology for each zone was assessed by visual inspection and was typified as recorded in Figure 3. These findings represent an 'impression' rather than an absolute or accurate rendition of the bone.
The angle of deviation between the sternoclavicular end and the shaft (SC/S∞), and the angle between the acromio-clavicular end and the shaft (AC/S∞) were captured by digital photographs and measurement by protractor (Figure 3).
The clavicular bow as measured in the transverse plane was measured directly from the clavicles with a protractor (Figure 4).
The results were recorded on a spreadsheet for ease of analysis. Minimum and maximum values for length and the cross-sectional dimensions are shown in Table I.
There were 44 left and 56 right clavicles. Age and gender were not known.
Length of clavicle
The mean length of the clavicles assessed was 150.1 mm with a range of 121.0-183.0 mm. The right was 151.6 mm (129 mm-174 mm) and the left side was 148.2 mm (121 mm-183 mm).
Length of each zone
The values for each 'third' of the clavicle was determined by simple mathematical division of the overall length into 3 and is recorded in Table I.
The sternal portion of the clavicle varied from a circular to quadrangular or prismatic profile in cross-section. Zones I and II were found to have predominantly tubular, either faceted or ovoid. Only in two instances was the AP dimension appreciably less than the superior/inferior dimension. Zone III, the central third, tended to have a circular to ovoid profile becoming more flattened at its lateral extent. Zones IV and V, the acromial portion or outer third, became flattened with the narrower dimension being supero-inferior. Figure 5 illustrates the different profiles encountered, and graphically orientates the dimensions superiorly and inferiorly. Table II illustrates the prevalence for each profile type in the five zones.
The measurements of the minimum and maximum crosssectional dimensions of its narrowest and widest parts for each zone are recorded in Table III. The narrowest measurement was 6.8 mm and the widest 33 mm.
Medial and lateral axis deviations
The angle of deviation between the sternoclavicular end and the shaft (SC/S), and the angle between the outer acromial clavicular end and the shaft (AC/S) were measured and recorded in Table IV. The SC/S angle was 25.2º with an SD of 7.3º and the AC/S angle was 38.6º with an SD of 9.8º.
The clavicle was found to have an angulation directed superiorly in all but one case. The bow angle mean value was 13.5º, (0º to 29º) and a SD of 8.5º. A comparison with the measurements of Huang et al are shown in Table V.
Clavicle fractures are common and occur most frequently (80-85%) in the middle third. Although usually treated conservatively, there is increasing recognition of the morbidity associated with displaced fractures.11,12 The choice of the sites at which the clavicles were measured was determined by the desire to acquire information about those areas where a fixation device, particularly an external fixator could be applied with the least interference of the soft tissues attached to or around the clavicle. The surface markings caused by the attachments of muscles and ligaments were identified as the extent of such structures.
The overall length of the clavicle measured in this study (150.1 mm) by direct dry bone measurement differed from those published by Ahmad et al who recorded a mean length of 136.2 mm (range: 112.6-172.0 mm) in a radiographic study, and by Gumina et al who recorded a mean of 138 mm ± 12.3 (SD) and Galley et al 138.4 (113.8-167.6) in wet cadaver dissections. The results reported by Huang (145.0 mm) in a CT study (USA), that of Daruwalla et al (144.3 mm) in the 3D CT (UK) and that of Nalla et al13 (143.3 mm) (RSA), also dry bone measurement, were all marginally less than the Cape Town study but reasons for our increased average length are not clear. While gender has been shown to play a role, gender identity of the specimens in our group was not known. This difference may reflect a true difference between different regions in the world, although discrepancies due to the differing methods of measurement cannot be excluded. In all reported series there is wide variation demonstrated in the dimensions, which reflects size variation in the general population.5
It can be seen that the traditional practice of dividing the clavicle into 'thirds' does not correspond completely with anatomical parameters. Although such a division is reasonable in a clinical setting for fracture configurations, a more rational approach when considering implant design is to take into account the soft tissue attachments to the clavicle.
The dimensions of each zone have not been previously recorded and this has been done so here. We believe the zone categorisation used here to be relevant in order to define the ideal area for the attachment and design of a fixation device. Knowledge of the appropriate area for fixation and the length of bone available for application of a fixation device are vital to optimal design of an external fixation device. In this study we found that the average length available for application of a device is 40 to 60 mm, but laterally this may be limited to 40 to 55 mm as defined by zones I and II and zones IV and V respectively.
The cross-sectional profile will also influence the ease or difficulty with which an external fixation device can be securely applied. From the cross-sectional profile it can be seen that zones I and II, the medial end, can be consolidated into one group with the profile being overwhelmingly tubular or somewhat quadrilaterally faceted, lending it to a variety of fixation options and inclinations. At the other end, the lateral end is flattened supero-inferiorly and fixation is more intuitively restricted to perpendicular to the superior/ inferior surfaces. In its mid portion the clavicle undergoes a transition from a faceted surface to a more rounded or ovoid profile which favours anterior and superior placement of a fixation device.
The angular change that occurs across the length of the clavicle in the transverse and coronal planes must be accommodated by the implant used for fracture fixation. For an exoskeletal device sufficient inherent adjustability is required to accommodate the curves and in addition it should have the versatility for application on the right and left sides, if the methods for bone attachment on the medial and lateral sides differ in accordance with the bone morphology.
Although not measured in this study it is of interest to note the cortical thickness variation. In a study by Ahmad et al they reported the central third of the clavicle to have the thickest cortex. In this study the mean cortical thickness (mean 3.37 mm) was greatest at a point 60% from the sternal end with the mean thinnest cortex (1.37 and 1.15 mm) found at the extreme sternal and acromial ends of the bone respectively. The reason for not conducting these measurements in our study was that the primary objective was to examine and record the external morphological characteristics, and furthermore the gender and age of the clavicles were not known: both of which influence cortical thickness.6,9 Caution should be exercised when interpreting measurements from dry bones because change in dimensions may occur during preparation.10
The mean angle differences between the medial and lateral ends of the clavicle to the shaft were 25.2º and 35.8º for the medial and lateral angles respectively which differs somewhat from the findings of Huang. It is possible that the technique of measurement may have varied between the two studies because the precise reference points for these measurements are not mentioned in the Huang study. In the study by Huang there was no significant difference between the angles in males or females or between black or white subjects.
The morphometric values obtained from this study can assist in the mathematical calculation or finite element analysis of the stress and forces that are transmitted through the clavicle during various activities, such as lifting the arm, thereby assisting with device design and fixation method for each specific area of the clavicle.
A potential weakness of this study is the sample size of 100 clavicles only that may not represent the full spectrum of variations encountered in the clavicle. This applies in particular to the upper and lower ends of the spectrum.
The practical value of this study has been to provide physical data regarding the dimensions and morphology of the clavicle. There is wide variation in the dimensions and morphology of the clavicles assessed. It can be seen that attachment of a fixation device can be easily achieved to the medial end. The morphology of this region permits application in a variety of orientations and inclinations. Laterally however the narrowness of the supero-inferior dimension restricts options for application of fixation devices.
1. Schuind F, Pay-Pay E, Andrianne Y, Donkerwolcke M, Rasquin C, Burny. External fixation of the clavicle for fracture or non-union in adults. J Bone Joint Surg Am 1988 Jun;70(5):692-5. [ Links ]
2. Demiralp B, Atesalp AS, Sehirlioglu A, Yurttas Y, Tasatan E. Preliminary results of the use of Ilizarov fixation in clavicular non-union. Arch Orthop Trauma Surg 2006 Aug;126(6):401-5. [ Links ]
3. Tomiç S, Bumbasirevic M, Lesiç A, Bumbasireviç V. Modification of the Ilizarov external fixator for aseptic hypertrophic nonunion of the clavicle: an option for treatment. J Orthop Trauma 2006 Feb;20(2):122-8. [ Links ]
4. McCormick WF, Stewart JH, Greene H. Sexing clavicles using length and circumference measurements. Am J Forensic Med Pathol 1991 Jun;12(2):175-81. [ Links ]
5. Jit I, Singh S. Estimation of stature from clavicles. Indian J Med Res 1956 Jan;44(1):137-55. [ Links ]
6. Ahmad M, Trewhella MJ, Bayliss NC. Study to describe the morphology of a series of clavicles and the dimensions of its intramedullary canal. J Bone Joint Surg, 2000;88-B, Issue SUPP III:401. [ Links ]
7. Gumina S, Salvatore M, De Santis P, Orsina L, Postacchini F. Coracoclavicular joint: osteologic study of 1020 human clavicles. J Anat 2002 Dec;201(6):513-9. [ Links ]
8. Huang JI, Toogood P, Chen MR, Wilber JH, Cooperman DR. Clavicular anatomy and the applicability of pre-contoured plates. J Bone Joint Surg 2007;89(A):2260-5. [ Links ]
9. Daruwalla Z, Courtis P, Fitzpatrick C, Fitzpatrick D, Mullet H. Anatomic variation of the clavicle: a novel 3-dimensional study. Clin Anat 2010;23:199-209. [ Links ]
10. Galley I, Watts A, Bain G. The anatomic relationship of the axillary artery and vein to the clavicle: a cadeveric study. Journal of Shoulder and Elbow Surgery 2009 18(5):21-25. [ Links ]
11. Lazarides S, Zafiropoulos G. Conservative treatment of fractures at the middle third of the clavicle: the relevance of shortening and clinical outcome. J Shoulder Elbow Surg 2006 Mar-Apr;15(2):191-4. [ Links ]
12. Chan KY, Jupiter JB, Leffert RD, Marti R, Clavicle malunion. J Shoulder Elbow Surg 1999;8:287-90. [ Links ]
13. Nalla S, Asvat R. Incidence of the coracoclavicular joint in South African populations. J Anat 1995 June;186(Pt 3):645-9. [ Links ]
The content of the article is the sole work of the authors. No benefits of any form have been or are to be received from a commercial party related directly or indirectly to the subject of the article.